# RESOLUTION, CALIBRATION AND OPERATIONAL DETAILS OF THE UTMSL FIVE METER LITTROW SPECTROMETER

Please use this identifier to cite or link to this item: http://hdl.handle.net/1811/15887

Files Size Format View
1973-AA-07.jpg 211.7Kb JPEG image

 Title: RESOLUTION, CALIBRATION AND OPERATIONAL DETAILS OF THE UTMSL FIVE METER LITTROW SPECTROMETER Creators: Gailar, Norman M.; Blass, W. E.; Jennings, D. E.; Lovell, R. J.; Boyd, William J.; McElyea, Gerald W.; Jacobson, Harry C.; Moore, Thomas F.; Moore, Dana H. Issue Date: 1973 Publisher: Ohio State University Abstract: The University of Tennessee Molecular Spectroscopy Laboratory has successfully operated the five meter Littrow spectrometer in the one to five micrometer region with a resolution better than $0.020 cm^{-1}$. Designed to work in the one to 25 $\mu$m region, the monochromator is a Littrow type which double passes a $20 \times 40$ cm, 79/mm Bausch and Lamb grating blazed at $63^{\circ} 26^{\prime}$. In the six to $25/ \mu$m region, slits 12.5 cm tall will be used to improve through-put. Tests of the instrument in the 6-25 $\mu$m region are scheduled for mid-1973. Resolution test data will be presented for the 1-5 $\mu$m region. Calibration is optimized using a Michelson interferometer attached to the grating table. Calibration test data will be discussed. A real time data acquisition and instrument control system designed around a PDP-11 computer is an integral part of the spectrometric system. Advantages of such a system will be illustrated with typical spectra and preliminary analysis results. Description: Author Institution: Molecular Spectroscopy Laboratory, Department of Physics and Astronomy URI: http://hdl.handle.net/1811/15887 Other Identifiers: 1973-AA-07